|Model||TVP600 Volatile Phenol Analyzer|
|Monitoring Factor||Volatile phenol|
|Range||(0.05-80) mg/L expandable|
|Measurement Mode||Manual measurement, continuous measurement, periodic measurement, fixed-point measurement, automatic calibration measurement, standard solution verification|
|Measurement Period||30 minutes to 24 hours, with adjustable expansion|
|Zero Drift||± 5% F. S|
|Range Drift||± 5% F. S|
|MBTF||> 720 hours|
|Stored data||Five years of data storage|
|Communication interface||4-20mA, RS232, RS485, dry contact|
|Contour Dimension||505 * 395 * 671(mm)|
Principle and Application of Volatile Phenol Analyzer
Volatile phenol is a highly toxic substance, a certain amount of human intake will appear acute poisoning symptoms; long-term drinking of water contaminated with phenol, can cause headache, rash, pruritus, anemia and a variety of neurological symptoms. When the content of phenol in water was 0.1 – 0.2 mg/l, the fish meat had peculiar smell, and when the content was more than 5 mg/l, the fish died of poisoning.
Principle: in alkaline medium, in the presence of potassium ferricyanide, phenolic compounds in water react with 4-aminoantipyrine to form orange-red compounds, the absorbance of which is proportional to the content of volatile phenol in water, the content of volatile phenol in water was calculated by measuring the absorbance of the compound.
It is suitable for laboratory Analyzer or industrial production process water, sewage, urban domestic sewage, river, lake, sea and other surface water testing.